General Scanning Recommendations Print
Modified on: Fri, 5 Mar, 2021 at 7:15 AM
Disclaimer: The information herein is not intended to replace the product’s User Manual. Please consult the User Manual for comprehensive information and cautionary guidance about the product’s operation and use.
Before Each Use
Inspect the Vscan Air CL probe. Refer ‘Inspection’ for details.
After Each Use
Inspect the Vscan Air CL. Refer ‘Inspection’ for details.
- Clean the Vscan Air CL (see ‘Cleaning and disinfection’).
- If required, disinfect the Vscan Air CL(see ‘Cleaning and disinfection’).
Ensure that the Vscan Air CL is properly cleaned and disinfected after each use and before storing in the protective case.
Like most high frequency computing devices, the electronic components of Vscan Air CL will generate some heat while operating normally and as intended. Vscan Air CL is equipped with safety mechanisms which will automatically reduce computing speed (frame rate), and ultimately shut down the device, before any risk of overheating occurs. Vscan Air CL is verified to comply with harmonized safety standards under any operating condition (see ‘Environmental requirements for Vscan Air CL’ ). To help keeping the Vscan Air CL operating temperature at an optimal functional level, and to ensure longer scanning time with maximum frame rate, it is recommended to hold the Vscan Air CL such that there is good contact between the device and the hand.
Did you find it helpful?Send feedback